> PT-SR-24C - Double Door Pass-Thru, Hinged Models
PT-SR-24C - Double Door Pass-Thru, Hinged Models
Company : Consolidated Stills & Sterilizers
UMDC Code : 102416
All BSL-3 autoclaves must be sealed appropriately in the wall to prevent microbes from passing between the contained and non-contained rooms (double door, thru wall systems). A biological sealing flange or Bio-Seal provides separation and a positive seal between these two rooms. BSL-3 sterilizers typically incorporate a steel plate that is continuously welded to all four sides of the sterilizer. It extends out beyond the frame of the sterilizer to which the installer then attaches the barrier. The other side of the barrier is then secured to the surrounding walls and sealed to the wall.
By design, the biocontainment lab is under lower or negative air pressure relative to the hallway or adjacent lab. When the seal is properly installed, a gas tightness or leak test (Halogen Tightness Test) will be performed to ensure that the bio-seal will withstand the differential pressure while still maintaining a biological barrier.
Product URL : http://www.consteril.com/bsl-3-medium-series-doubl
Last Updated : 11/09/2012